Very Low FiO2 activates when the monitored FiO2 is less than 18%.
Follow these steps to troubleshoot this alarm:
- Ensure the VOCSN air inlets are not blocked and there is adequate air flow around the device.
- Remove any oxygen sources connected to VOCSN or the patient circuit.
- If the Very Low FiO2 alarm resolves, troubleshoot problems with the oxygen source.
- If the Very Low FiO2 persists, there may be a problem with the VOCSN oxygen sensor. Please fill out this form. One of our clinical specialists will contact you.